Bluetooth OLED Display Fingertip Pulse Oximeter Finger pulse Oximeter for HR and SPO2
This SP001 Fingertip Pulse Oximeter can be used in measuring the pulse oxygen saturation and pulse rate through finger. The product is suitable for being used in family, hospital, oxygen bar, community healthcare, physical care in sports (It can be used before or after doing sports, and it is not recommended to use the device during sports or exercising)
1.Integrated with SpO2 probe and processing display module
2.Small in volume, light in weight and convenient in carrying
3.Operation of the product is simple, low power consumption
4.Operation menu for the function setting
5.SpO2 value display
6.Pulse rate value display, bar graph display
7.Pulse waveform display
8.A pulse rate sound indication
9.With measured data overruns limits and low-voltage alarm function, the upper/down alarm range can be adjustable
10.Low-voltage indication: Low-voltage indicator appears before working abnormally which is due to low-voltage, and with alarm function
11.With SpO2 value and pulse rate value of storage, the storage data can be uploaded.
12.RF Wireless communication function
13.Connected with an external oximeter probe
14.Automatically power off function: When the device is under the state of measuring interface. It will automatically power off within 10 seconds if the finger falls out of probe.
Description: Fingertip Pulse Oximeter for SPO2 and pulse rate with Bluetooth connectivity.
Display Mode: Color OLED display
Screen Resolution: 128*64
SpO2 Measuring Range: 0%~100%(the resolution is 1%)
Accuracy: 70%~100%: ? 2%, Below 70% unspecified
PR Measuring Range: 30bpm~250bpm, (the resolution is 1bpm)
Accuracy: ? 2bpm or ? 2% (select larger)
Resistance to surrounding light: The deviation between the value measured in the condition of man-made light or indoor natural light and that of darkroom is less than ? 1%.
Power Consumption: Less than 80mA
Voltage: DC 2.6V~3.6V
Power Supply: 1.5V (AAA size) alkaline batteries ? 2
Battery working time: The theoretical number is 32 hours.
Safety Type: Interior Battery, BF Type
Measurement Performance in Weak
Filling Condition: SpO2 and pulse rate can be shown correctly when pulse-filling ratio is 0.4%. SpO2 error is ? 4%, pulse rate error is ? 2 bpm or ? 2% (select larger).